Gel Permeation Chromatography (GPC) is a separation method for the determination of molecular weight averages (Mn) and molecular weight distribution (PDI=Mw/Mn). The testing report can include calibration plot, chromatogram, number average molecular weight and PDI of the polymer. GPC-DMF is using DMF as a solvent. Gel Permeation Chromatography (GPC) separates molecules on the basis of hydrodynamic size (not Mw). The large molecules move more rapidly through the GPC column, and in this way the mixture can be separated. The elution volume of a molecule is related to its hydrodynamic volume. To relate the elution volume to the molecular weight of a polymer, a calibration with molecular weight standards is needed.
Pump: Waters Alliance HPLC System, 2695 Separation Module
Detectors: Waters 2414 Differential Refractometer (RI) and Waters 2998 Photodiode Array Detector (PDA)
Solvent: DMF containing 0.01% of LiBr
Flow Rate: 0.3 mL/min
Columns for GPC-DMF:
2 Tosoh TSKgel Super HM-M columns
Solvent: DMF with 0.01% LiBr
The addition of the electrolytic salt, LiBr (0.01%), stabilizes the effects whether associated with the polymer or the columns, enabling normal fractionation to occur.
Calibration: vs polystyrene standards, per request vs PEO or PMMA.
MRL room 1043
43 Maple Street
Milford, MA 01757
Internet - http://www.waters.com
Instrument Information and/or Manual:
|GPC DMF Quick Start Guide||126.03 KB|
Please contact Rachel Behrens (rachel [at] mrl [dot] ucsb [dot] edu) to schedule training. Before training starts please read the MANUAL.